Readers of this blog know that I’m in favor of specifications that account for 100% of the results. The danger of specifications that are for 95% or 99% of the results is that errors can occur that cause serious patient harm for assays that meet specifications! Large and harmful errors are rare and certainly less than 1%. But hospitals might not want specifications that account for 100% of results (and remember that hospital clinical chemists populate standards committees). A potential reason is that if a large error occurs, the 95% or 99% specification can be an advantage for a hospital if there is a lawsuit.
I’m thinking of an example where I was an expert witness. Of course, I can’t go into the details but this was a case where there was a large error, the patient was harmed, and the hospital lab was clearly at fault. (In this case it was a user error). The hospital lab’s defense was that they followed all procedures and met all standards, e.g., sorry but stuff happens.
As for irrelevant statistics, I’ve heard two well-known people in the area of diabetes (Dr. David B Sachs and Dr. Andreas Pfützner) say in public meetings that one should not specify glucose meter performance for 100% of the results because one can never prove that the number of large errors is zero.
That one can never prove that the number of large errors is zero is true but this does not mean one should abandon a specification for 100% of the results.
Here, I’m reminded of blood gas. For blood gas, obtaining a result is critical. Hospital labs realize that blood gas instruments can break down and fail to produce a result. Since this is unacceptable, one can calculate the failure rate and reduce the risk of no result with redundancy (meaning using multiple instruments). No matter how many instruments are used, the possibility that all instruments will fail at the same time is not zero!
A final problem with not specifying 100% of the results is that it may cause labs to not put that much thought into procedures to minimize the risk of large errors.
And in industry (at least at Ciba-Corning) we always had specifications for 100% of the results, as did the original version of the CLSI total error document, EP21-A (this was dropped in the A2 version).